Mumi A further point are inconsistencies between the online help, the help function in the Spider setup program and the printed manual! My OS is windows 98, labview version 6i. I have 2 examples and one shows the mesured value splder8 the port and the other plots graph in the screen of the signal mesured in the port. The cable has got a length of about 1 meter. Message 5 of I tried to run one of the spider8 examples with labview and did not work. Furthermore, the system is very unstable.
|Published (Last):||1 February 2012|
|PDF File Size:||9.55 Mb|
|ePub File Size:||2.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Samura So in this example, the timeout has to be adjusted in respect to the measuring rate. It seems as the hbj is not fast enough. I was not able to find much in the way of documentation on the HBM website regarding the drivers you mentioned. Message 8 of Spider with the P22 firmware version. I would like to receive help with respect this issue. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. First I chose a baud rate ofnow I use This Tutorial has bbm regarding how to properly call functions from DLLs using the call library node.
A further point are inconsistencies between the online help, the help function in the Spider setup program and the printed manual! Last but not least the timeout spided8 to be adjusted according to the specific number of counts, baud rate and measuring rate.
W henever this is not true, the Spider should be turned off and on. The big advantage is, that you have all options to control your DAQ process. There is one little disadvantage: Message 1 of The cable has got a length of about 1 meter. For Labview there are some examples which show how to access data via catman activeX. This is not very useful! Please could you help me to know what the correct steps are to try the spider89 works with Labview?.
I have read that minimum version shoud be P20 to work correcrtly with Labview. I have updated everything. The Spider is connected with the notebook via the serial port. I solved the problems a few days ago.
Can anybody help me? Best Regards Andreas R. First of all, my operating system is NT4. My OS is windows 98, labview version 6i. I spiedr8 2 examples and one shows the mesured value in the port and the other plots graph in the screen of the signal mesured in the port. If you need the archives email-me wanderleyff gmail. Also, which drivers are you using exactly? The Spider 8 is now equipped with hm latest firmware, which can be obtained from the HBM homepage.
I hope this help you. Soon I will tell what were the problems. Most Active Software Boards: May be, he has a demo version for you almost catman professional including the catman activeX interface. I downloaded the LabView driver for this measurement device from the NI homepage.
Most 10 Related.
USB-Adapter-HBM Spider8 / CP32 Drivers
HBM SPIDER8 PDF